NeuLog Force Logger Sensor
Connect the sensor to a voltage source (NeuLog's Bridge, Display Unit or Battery Unit). To zero the reading, just press the sensor's push-button for about 3 seconds.
*Requires connection hardware, sold separately.
Experiment Duration: 50 milliseconds to 31 days.
Range | ADC Resolution | Resolution | Max Sample Rate (S/sec):
+/- 10 N, 10 bit, 0.01 N, 3000
+/- 50 N, 10 bit, 0.01 N, 3000
|We don't have that information available online. If you'd like to know, please email us at firstname.lastname@example.org and we'll get you that information ASAP.|
|Instructions||No instructions available online. Please contact us for further assistance.|
|High Resolution Image||Please register as a teacher or distributor to view and download high resolution images.|