NeuLog pH Logger Sensor
This sensor gives a fast response across the full pH range and can be calibrated with any standard buffer solution. The sensor should be connected to a voltage source (the USB module plugged into a PC, Neulog's™ Display Unit or Battery Unit). Insert the sensor into a pH = 7 buffer and press the sensor's push button for about 3 seconds. The reading is calibrated to 7.
*Requires connection hardware, sold separately.
Experiment Duration: 1 second to 31 days.
Range | ADC Resolution | Resolution | Max Sample Rate (S/sec):
0 to 14, 16 bit, 0.01pH, 100
|We don't have that information available online. If you'd like to know, please email us at email@example.com and we'll get you that information ASAP.|
|Instructions||No instructions available online. Please contact us for further assistance.|
|High Resolution Image||Please register as a teacher or distributor to view and download high resolution images.|